A New Advanced Mercury Analyser from CKIC
Jan 26 2017 Comments 0
Mercury is the most potent neurotoxin known to man. Nothing affects your brain more drastically and critically, mercury doesn't break down in the body or in the environment. It just keeps accumulating.
So, it’s vital to monitor potential sources of mercury and reduce people’s exposure wherever possible.
One of the easiest and best ways to do that is by the use of CKIC’s new mercury analyser.
The 5E-HGT2320 will directly determine total mercury in materials such as Coal, Oil, Soil, Food, Feed, Fertiliser, Ore, Wastewater and Sludge.
Samples are introduced directly into the instrument and the result is obtained in five minutes. There is no need for preparation saving time and money over traditional methods.
Operation couldn’t be easier. The sample is placed into a boat and weighed. Powerful software takes over and completes the test automatically. Up to 70 samples can be loaded and analysed sequentially.
5E-HGT2320 employs the thermal decomposition process described in U.S. EPA Method 7473.
The operating sequence is as follows.
Firstly the samples are dried and the released moisture is discharged in the form of vapor through a three-way valve. The sample is then thermally decomposed in an oxygen environment and the decomposition products are carried by oxygen flow to the catalytic section of the furnace, where oxidation is completed and all interfering substances, such as halogens and oxides of nitrogen and sulfur are eliminated. The remaining decomposition products are then carried into a gold amalgamator where the mercury is selectively trapped.
The system continues to be flushed with oxygen until the remaining decomposition products are removed from the system and then the amalgamator is rapidly heated releasing the trapped mercury. The vapour is swept through a 253.7nm wavelength atomic absorption spectrophotometer and mercury concentration is measured via absorbance peak height or peak area.
Sample weights of up to 1.5g can be run with a detection limit of 0.001 ng Hg.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This edition.. Business News Gas Detection - Growing Market for CEMS in Developing Countries Air Monitoring - Mistakes and Problems in Gas Sampling and Conditioning - ECO PHYSICS...
View all digital editions
Nov 23 2017 Brussels, Belgium
Nov 29 2017 Antwerp, Belgium
Dec 04 2017 MUSCAT, OMAN
Dec 12 2017 Nuremberg, Germany
Jan 09 2018 Leeds, UK