The importance of bi-directional compressed air flow measurement
Oct 13 2020 Read 377 Times
Bi-directional flow occurs more often than you would expect in compressed air systems. In this article you will find a few examples of bi-directional flow situations. In these cases, a bi-directional flow meter is the only way to measure correctly and to know what is going on in your compressed air system.
Example case: Ring network with two compressor rooms
A large electronic component manufacturer has a compressed air system that consists of a ring network with air supplied by two compressor rooms (see picture). In compressor room B two flow meters were installed after the receiver tank, due to space restrictions. At the beginning, these were uni-directional flow meters. In other words, they could not see the difference between forward and backward flow.
Receiver tank acts as big balloon
After a couple of days strange measurements were observed. The flow meters, manufactured by VP Instruments, were showing considerable measurement values when the compressor in room B was turned off. Measurements with VPFlowScope bi-directional flow meters showed that the large receiver tank is filled by the compressors from the other compressor room when the local air demand in the ring network is low. Moreover, it delivers air to the network when the local demand is high. You can see the receiver tank as a big balloon interacting with the network.
“Merging traffic” situation
Furthermore, when the compressor in room B is running loaded, compressed air is fed into the ring network. In the T-junction, the flow merges with existing “traffic” based on how the consumption in the ring mains is balanced. If the consumption is well balanced, it might be perfectly symmetric (50% left, 50% right), but when the consumption on one side is much higher, the air flow will distribute differently. The air coming from the compressor merges with the air which was already flowing through the ring network. This can result in a high or low reading, depending on the actual demand and the resulting flow direction.
Other bi-directional flow examples
Bi-directional flow meters can reveal important issues in many compressed air systems. Other examples where you can reveal issues:
- Leaking non-return valves just after the compressor, or leaking condensate drains. The bi-directional flow meter will tell you when it happens.
- Overseen branches
- Systems with multiple compressor rooms
- Decentralized receivers can also act as a local buffer
- Oscillations in the pipe network can be seen as consumption instead of near zero flow
Uni-directional flow meters will show a positive flow when reverse flow occurs. This can lead to wrong conclusions about the system behavior. Discover your actual consumption and avoid such misreading with bi-directional flow meters, like the VPFlowScope. It is recommended in ring networks to always use bi-directional flow meters. In case of doubt, just use a bi-directional flow meter. Costs for this additional feature in your flow meter is insignificant compared to drawing the wrong conclusions.
Read the entire article with more examples and details here.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Business News - Global valve event to incorporate technology in the battle against coronavirus - New ASTM International standard will help test water quality - Acquisition t...
View all digital editions
Oct 26 2020 Virtual event
Oct 27 2020 Moscow, Russia
Oct 27 2020 Kiev, Ukraine
Oct 27 2020 Moscow, Russia
Oct 28 2020 Beijing, China