Eliminate the Impact of Instrument-Related PFAS Interferences by Using a Delay Column
Mar 08 2022
Author: Restek on behalf of Restek Corp
Free to read
This article has been unlocked and is ready to read.
Poly- and perfluoroalkyl substances, or PFAS, are rapidly emerging as some of the most important environmental contaminants to monitor around the world. Their widespread use and environmental persistence make them truly a global issue. Concerns over possible health risks are driving environmental scientists to look for these compounds everywhere from penguins in the south to polar bears in the north, and many points in between, and they are discovering the extent of PFAS contamination is literally pole to pole. However, one other concerning place where at least some of these compounds are present is inside the very instruments used for PFAS analysis of environmental samples. Background PFAS contamination can interfere with low-level quantification requirements, which are usually in the range of parts per trillion (ppt) or below. Instrument-related background contamination can stem from LC components that contain fluoropolymers (e.g., PTFE), which can leach into mobile phase solutions. Using tubing and containers without fluorochemicals and switching instrument components to PEEK or stainless steel mitigates some of the risk of background contamination, but it does not eliminate the impact of interferences on trace-level analysis, so another solution is required.
In a standard setup, instrument-related PFAS interferences move through the system with the mobile phase and collect at the head of the analytical column between sample injections. However, adding a second column, called a “delay column,” into the mobile phase flow path immediately before the injector is an effective way catch the background PFAS before they reach the analytical column (Figure 1). In this scenario, when the sample is injected, only PFAS in the sample will focus at the head of the analytical column. Then, during the course of a gradient analysis, PFAS from the sample will begin to move through the analytical column like normal. The background PFAS trapped in the delay column during equilibration will be eluted and pass through the analytical column as well. However, the background PFAS will elute later than the sample PFAS because they entered the analytical column after the injected sample due to additional retention by the delay column prior to reaching the injector. In addition, the background PFAS never focus on the head of the analytical column because they are present in the flow path in continuum rather than as a discrete injection. The net effect is that by using a delay column you can differentiate instrument-related PFAS background contamination from target PFAS in a sample.
Figure 2 is an example of the kind of background contamination that interferes with accurate PFAS analysis. The system was equilibrated for 10 minutes without a delay column to allow instrument-related PFAS interferences to accumulate at the head of the analytical column before a blank injection. In this case, five potential target compounds were observed in the instrument blank. Which compounds are present and to what degree they are observed ultimately depends on the specific analytical setup. System-related PFAS contamination can be affected by factors such as instrument make and model, tubing and fitting materials, mobile phase solvent grade, and analytical method conditions.
Adding a delay column to the flow path is an effective way to retain background-related PFAS interferences so they do not coelute with PFAS from the sample during analysis. When the experiment in Figure 2 was repeated with a delay column installed, there were no observable peaks in the blank injection at the retention time for each PFAS previously found, even after relatively long equilibration times. Instead, the background PFAS interferences from the instrument eluted in a wide band after the target retention time. Because band broadening occurs while the background PFAS compounds are being retained by the delay column during equilibration, the signal for background PFAS does not appear as a well-defined peak once the background PFAS elute. Instead, the background signal is observed as a region of slightly elevated baseline (Figure 4).
Since the delay column adds additional length to the flow path between the mixer and the injector, target analyte retention times also shift slightly because of the longer time it takes for mobile phase changes to reach the analytical column during a gradient analysis (Figure 3). The net effect of a PFAS delay column is demonstrated in Figure 4: target analyte retention times are distinct from the instrument-related PFAS elution region, so there is no interference from the instrument background during sample analysis. Note that the elution time regions for instrument-related PFAS shown in Figure 4 were determined by lengthening the equilibration times and observing the baseline deflection caused by a broad band of delayed instrument-related PFAS eluting from the analytical column.
In summary, using a PFAS delay column can eliminate the detrimental impact of background PFAS interferences from instrument-related sources by retaining contaminants prior to the analytical column and eluting them only after the sample has been injected and the gradient elution has started. The elution delay between the target PFAS in the sample and the background PFAS from the LC system is enough to separate them sufficiently to allow for accurate quantification of the target compound in the sample.
 A. Schiavone, S. Corsolini, K. Kannan, L. Tao, W. Trivelpiece, et. al. Perfluorinated contaminants in fur seal pups and penguin eggs from South Shetland, Antarctica, Sci. Total., 407 (12) (2009) 3899-904. https://doi.org/10.1016/j.scitotenv.2008.12.058
 S. Tartu, S. Bourgeon, J. Aars, M. Andersen, et al. Diet and metabolic state are the main factors determining concentrations of perfluoroalkyl substances in female polar bears from Svalbard, Environ. Pollut., 229 (2017) 146-158. https://doi.org/10.1016/j.envpol.2017.04.100
 K. Rankin, S. Mabury, T, Jenkins, J. Washington, A North American and global survey of perfluoroalkyl substances in surface soils: Distribution patterns and mode of occurrence, Chemosphere, 161 (2016) 333-41. https://doi.org/10.1016/j.chemosphere.2016.06.109
 B. Gonzalez-Gaya, J. Dachs, J Roscales, G. Cabellero, B. Jiminez, Perfluoroalkylated Substances in the Global Tropical and Subtropical Surface Oceans, Environ. Sci. Technol., 48 (22) (2014) 13076-84. https://pubs.acs.org/doi/abs/10.1021/es503490z
In This Edition Water/Wastewater - Continuous remote water quality monitoring networks Environmental Laboratory - The Important Role of ICP-MS in Understanding the Toxicological Link Be...
View all digital editions
Feb 28 2024 Chennai, India
Feb 29 2024 Dhaka, Bangladesh
Feb 29 2024 Kathmandu, Nepal
Mar 04 2024 Mumbai, India
Mar 04 2024 Guanghzou, China