9 Tips to select & install compressed air flow meters correctly
Feb 12 2020 Read 1257 Times
Prevent wrong conclusions by incorrect measurements. Flow meters are great tools to measure and manage energy consumption, only though when selected and installed properly. In this [eBook] we will walk you through the major criteria for proper compressed air flow measurement.
Here we present a compact overview. Download here the complete eBook for all information; including a nice checklist.
1. Select the right type flow meter for your application
Is the air wet or dry? Is an In-line or insertion probe style best for your application? Select the right technology and model to match your situation.
2. Make sure the air quality matches the flow meter
Regardless of type, any flow meter is sensitive to pollutants in the air. E.g. we recommend placing thermal mass flow meters in dry air downstream of all air treatment filters to prevent sensor degradation. Also, for differential pressure flow meters a combination of too much water and dirt can clog the sensor.
3. Create enough straight pipe length
A symmetric, fully developed turbulent flow profile is what is needed for accurate flow measurement. Install your flow meter at the recommended distance from bends, valves and other objects that can distort the flow profile. Check out here our piping table with some examples.
4. Beware of temperature sweeps
Avoid large and rapid temperature variations, e.g. downstream a malfunctioning desiccant dryer. This can affect the measurement accuracy of the flow meter, and extreme temperatures can also shorten its service lifetime.
5. Make sure the compressed air flow meter is 3-in-1
Get the complete picture and always measure mass flow, pressure and temperature at once. Our VPFlowScope flow meters all measure mass flow, pressure and temperature simultaneously.
6. Be aware of reverse flow conditions
Flow in compressed air pipes can reverse direction unexpectedly and only a bi-directional flow meter can reveal this. Reverse flow can reveal issues with ring networks, multiple compressor rooms, non-return valves, leakage inside compressors, drains, filters and dryers.
7. Understand the effect of a receiver
Measuring flow and pressure before or after the receiver tank makes a big difference! If you do not take the receiver into account, you can draw the wrong conclusions about compressor capacity or equipment capacity.
8. Wet air flow meter: prevent flooding or excessive water
Measuring wet air in the discharge pipe of a compressor is possible with a differential pressure flow meter; but be aware of excessive water. Check the water content of the system before installing your flow meter and check the flowmeter a short time after installation.
9. Wet air flow meter: install under the right angle
Flow meters for saturated air measurement need to be able to drain excess water build-up and gravity helps. Install the flow meter at a 30 to 45-degree angle in a horizontal pipe so water can easily drop off the meter’s tip.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Business News - Bavaria combines forces in hydrogen research - Vaisala appoints Kai Öistämö as President and CEO - Jean-Philippe Amiet appointed Global Sales Manager of C...
View all digital editions
Aug 11 2020 Guanghzou, China
Aug 12 2020 Auckland, New Zealand
Aug 13 2020 Shanghai, China
Aug 16 2020 Guangzhou, China
Aug 19 2020 Mumbai, India